Cellix has developed a novel microfluidic platform consisting of a PC- controlled Mirus Evo pump with microfluidic biochips. This enables very accurate flow rates to be achieved which are more reproducible and consistent compared to anything currently available. Importantly, flow rates are extremely low (5 pL/min to 10 μL/min) and the shear stress levels that the pump can mimic (up to 30 dyne/cm2) are equivalent to those found in blood vessels in vivo. The Vena8 Fluoro+ biochips are comprised of eight microcapillaries and are manufactured such that the dimensions of the capillaries are similar to the blood vessels being assessed. Currently, the capillaries may be coated with recombinant human adhesion proteins for use in inflammation studies. The Mirus Evo is vital to the use of small diameter capillaries as standard syringe pumps are incapable of delivering the required low flow rates.
First of all, the cell type to be analysed must be determined. This is followed by establishing how to harvest such cells e.g. culturing in growth media, or isolation from in vivo fluids. Secondly, the assay itself should be outlined, including what proteins will be used to coat the capillaries of the biochip. Thirdly, the adhesion profile of the cells to be passed through the coated capillary should be determined. Next, if exogenous compounds are being analysed, these should then be introduced to the system and their effect on the adhesion profile assessed. This should include calculation of required concentrations and pre-incubation conditions, before introduction to the system. Finally, the images taken via the digital camera attached to the microscope should be masked and analysed using the Image Pro Premier software.
Choice of Cell Type and Harvesting Protocols
A microfluidic assay assessing the effect of levocetirizine on human eosinophil adhesion was developed. Human eosinophils were obtained from blood donated by normal volunteers, or individuals with mild allergic disease who were not taking any medication and who gave informed consent (eosinophilia <0.5 x 10^6 eosinophils/mL).
Assay Outline, Including Vena8 Biochip Coating Procedures
Vena8 Fluoro+ biochip was coated for 1 h in humid conditions at ambient temperature with either rhVCAM-1 or BSA (both 10 μg/mL in HBSS containing Ca2+ and Mg2+). All capillaries were then coated with BSA to occupy non-specific binding sites. Resting or GM-CSF-treated eosinophils were pre-incubated at 37°C in a water bath for 10 mins before incubation with/without levocetirizine (0.1–100 nM), with anti-VLA-4 mAb as a positive control) for a further 20 mins.
Eosinophils were infused into the capillaries at stepwise increases in shear stress, from 0 to 5 dyne/cm2, 1 min per shear stress level. Images at each shear stress level were captured using the accompanying PixeLINK microscopy software. For experiments with GM-CSF- stimulated (1 ng/mL) eosinophils, the cytokine was added to the warmed cells at the same time as levocetirizine and incubated at 37°C for 20 mins prior to commencing the flow assay. Adhesion was evaluated by monitoring eosinophil migratory behaviour in real time with images captured via a digital camera connected to the microscope using VenaFlux assay software.
Three images per shear stress level were captured (see figure 1 for examples), and adhered eosinophil numbers were recorded using Image Pro Premier software. Data was exported into Excel for interpretation. Statistical significance was determined by Students unpaired t-test, and P>0.05 was considered statistically significant. Data was presented as mean±SEM.
We examined the effect of a concentration range of levocetirizine on GM-CSF- stimulated (10 ng/mL — optimal concentration established in preliminary experiments) eosinophil adhesion under flow to rhVCAM-1. Pre-incubation of eosinophils with GM-CSF increased the adhesion level by more than 50%, and also rapidly generated significantly more flattened cells at both the 5 and 15 min time-points. Levocetirizine significantly inhibited GM-CSF stimulated eosinophil adhesion to VCAM-1 and the ability of eosinophils to flatten on rhVCAM-1, in a dose-dependent manner, at both 5 min and 15 min. The data are summarized in Figure 2, with optimal inhibition of adhesion observed at a levocetirizine concentration of 10-8 M with an EC50 of 10- 9 M. However, a greater effect on GM-CSF- dependent eosinophil flattening was observed at 15 min. Pre-treatment of GM- CSF-stimulated eosinophils with a CD11b mAb had no measurable effect (data not shown).
The following solutions were used during the above procedure:
Eosinophil Isolation Media
HBSS without Ca2+ and Mg2+
100 U penicillin/0.1 mgstreptomycin
Note: incubate eosinophils at 5 x 10^6/mL in the adhesion media for at least 30 mins before commencing the assay. Also, dissolve eosinophil stimulants/inhibitors in this media.
Blaylock, M.G., Sexton, D.W., & Walsh, G.M (1999). Ligation of CD45 and the isoforms CD45RA and CD45RB accelerates the rate of constitutive apoptosis in human eosinophils. J Allergy Clin Immunol, 104, 1244-50.
Wu, P., Mitchell, S., & Walsh, G.M. (2005). A new antihistamine levocetirizine inhibits eosinophil adhesion to vascular cell adhesion molecule-1 under flow conditions. Clin. Exper. Allergy, 35, 1073- 1079.
Williams, V., Kashanin, D., Shvets, I.V., Mitchell, S., Volkov, Y., & Kelleher, D. (2002). Microfluidic enabling platform for cell-based assays. J. Assoc. Lab. Automation, 7, 135-141.
©2018 Cellix Limited. All rights reserved.